Pedal size: (W) 400mm*(L) 450mm*(H) 20mm
Left shoe setting value: 750KΩ—100MΩ
Right shoe setting value: 750KΩ—100MΩ
Adjustable distance of movable bracket: 900mm～1350mm
Wrist strap setting value: 750KΩ—10MΩ
Total weight: 6.5kg
Test accuracy: +-10%
Remark: This parameter is the GB standard. The conductive shoes are 100KΩ-35MKΩ, which can be calibrated according to the user's use conditions.
Wrist strap/footwear tester is a device designed to measure the comprehensive resistance of the human body (including the resistance of the human body, wrist straps and shoes, etc.). This instrument can detect and alarm when the comprehensive resistance of human body is more than 10E9 and less than 10E5.
Wrist strap/footwear tester is formulated with 750KΩ-10MΩ circuit (for testing wrist straps) and 750KΩ-100MΩ circuit (for testing shoes), this instrument is ideal for testing wrist straps and shoe covers.
1. This product can quickly test ESD equipment, and three independent devices (wrist strap, left shoe, right shoe) can complete ESD testing at one time.
2. There are three state displays for each of the three, LO (below the set value), PASS (within the set value range), and HI (higher than the set value) have an alarm function.
How to operate?
1.Turn the battery switch to the ON position before use. If the display is abnormal or the buzzer beeps for a long time, it means that the battery in the machine is low, please insert the battery adapter to charge it, and the machine can be used directly. The quiescent DC current is 12V/mA.
2.During the ESD test, as long as both feet are on the opposite metal pedals, the wrist strap is worn or inserted into the wrist strap connection port, and the finger is lightly pressed on the metal test button to know at a glance whether the ESD equipment is qualified.
Note: Do not wear gloves or use your fingernails to press the metal test button.